Impact of tissue heterogeneity correction on Gamma knife stereotactic radiosurgery of acoustic neuromas
Gabrielle W Peters, Christopher J Tien, Veronica Chiang, James Yu, James E. Hansen and Sanjay Aneja
Purpose/objectives: Treatment planning systems (TPS) for Gamma Knife stereotactic radiosurgery (GK-SRS) include TMR10 algorithms, which assumes tissue homogeneity equivalent to water, and collapsed-cone convolutional (CCC) algorithms, which accounts for tissue inhomogeneity. This study investigated dosimetric differences between TMR10 and CCC TPS for acoustic neuromas (ANs) treated with GK-SRS.
Materials/methods: A retrospective review of 56 AN treated with GK-SRS was performed. All patients underwent MRI and CT imaging during their initial treatment and were planned using TMR10. Each plan was recalculated with CCC using electron density extracted from CT. Parameters of interest included Dmax, Dmin, D50%, cochlea Dmax, mean cochlea dose, target size, and laterality (>20 mm from central axis).
Results: Median target volume of patients was 1.5 cc (0.3 cc-2.8 cc) with median dose of 12 Gy prescribed to the 50% isodose line. Compared to CCC algorithms, the TMR10 calculated dose was higher: Dmax was higher by an average 6.2% (p < 0.001), Dmin was higher by an average 3.1% (p < 0.032), D50% was higher by an average of 11.3%. For lateralized targets, calculated Dmax and D50% were higher by 7.1% (p < 0.001) and 10.6% (p < 0.001), respectively. For targets <1 cc, Dmax and D50% were higher by 8.9% (p ≤ 0.009) and 12.1% (p ≤ 0.001), respectively. Cochlea Dmax was higher, by an average of 20.1% (p < 0.001).
Conclusion: There was a statistically significant dosimetric differences observed between TMR10 and CCC algorithms for AN GK-SRS, particularly in small and lateralized ANs. It may be important to note these differences when relating GK-SRS with standard heterogeneity-corrected SRS regimens.
Keywords: Gamma Knife, acoustic neuroma, radiosurgery, treatment planning
After payment has been processed for your order of a digital copy (PDF) of this article, you will see a download link on your completed order page and also receive an email containing a download link. The links, which will enable you to download one copy of the article, will expire after 24 hours.